BOCON sterilize medical equipment has been specially designed for sterilization of products sensitive to high temperature and to humidity (Syringes, Defluxers, Catheters, Cartridges for Dialysis, Plastic articles, Bandages, Sutures etc.).Send Inquiry
BOCON sterilize medical equipment has been specially designed for sterilization of products sensitive to high temperature and to humidity (Syringes, Defluxers, Catheters, Cartridges for Dialysis, Plastic articles, Bandages, Sutures etc.).
Selection sealing door
door sealing way: air feeded inside.
|Selection control system cabinet|
Sterilizer Maintenance Guide
1. In accordance with the relevant provisions of the National Metrology Law, sensors and meters should be sent to the local metrology department for verification.
Fifteen, replace the circulating water of the heating system every six months.
2. According to the frequency of use of the sterilizer, replace the sealing door seal ring at least once every six months.
3. Check the sealing condition of the air valve once a week. If there is air leakage, tighten or change the sealing ring.
4. Before each sterilization, check whether the decompression value of the pressure reducing valve is correct. If there is any error, adjust it in time.
Nineteen, dredge the drug delivery line once a year.
5. Depending on the quality of the sterilant, the ethylene oxide filter should be cleaned regularly.
6. Clean or replace the air filter regularly according to the actual conditions of the working environment.
7. Regularly clean the dirt from the mouth of the ethylene oxide in the cabinet.
Twenty-three, the water ring vacuum pump should be supplied with water while the vacuum pump is starting, otherwise the vacuum pump will be damaged.
Twenty-four, steam generator, please refer to the instruction manual that comes with the manufacturer, and use it correctly.